Please use this identifier to cite or link to this item:
WRRCTR No. 67 Rainfall Intensity Instruments and Measurements
|Title:||WRRCTR No. 67 Rainfall Intensity Instruments and Measurements|
|Authors:||Fullerton, Charles M.|
Raymond, David J.
|LC Subject Headings:||Rainfall frequencies -- Hawaii.|
Rain and rainfall -- Hawaii.
|Date Issued:||Jul 1973|
|Publisher:||Water Resources Research Center, University of Hawaii at Manoa|
|Citation:||Fullerton CM, Raymond DJ. 1973. Rainfall intensity instruments and measurements. Honolulu (HI): Water Resources Research Center, University of Hawaii at Manoa. WRRC technical report, 67.|
|Series:||WRRC Technical Reports|
|Abstract:||Two instruments have been used to measure the rate of rainfall from tropical clouds over the windward coast of the Island of Hawaii. The Workman intensity gauge employs strain gauges to sense the deflection in a movable trough and hence the flow of water down the trough. The Raymond-Wilson gauge, a new instrument developed during this program, measures the electrical resistance of flowing water to determine the rainfall rate. These instruments were deployed in a small scale array of eight gauges to measure space and time variations in high intensity rainfall as showers move across the network. Comparative measurements were made with a standard weighing rain gauge, a tipping bucket rain gauge and the Jardi rate of rainfall gauge. Both the Workman gauge and the Raymond-Wilson gauge are clearly superior to the other instruments, because of their rapid response time and their capacity to measure high intensity showers continuously and automatically. The Raymond-Wilson gauge contains no moving parts and is operationally more suitable than the Workman gauge for deployment in a network of gauges. A sensor which measures the rate of change of atmospheric pressure has also been developed and is being used in the research program. An example of rainfall intensity measured during a 43-day period in October and November, 1972, is presented, along with a preliminary estimate of the in-cloud liquid water content of shallow Hawaiian clouds.|
|Pages/Duration:||vi + 32 pages|
|Appears in Collections:||
WRRC Project Reports|
Please email firstname.lastname@example.org if you need this content in ADA-compliant format.
Items in ScholarSpace are protected by copyright, with all rights reserved, unless otherwise indicated.